High quality radar level measurement manufacturer? So what are the installation technical requirements for radar water level meters? The installation of the radar water level meter must be vertical to the object to be measured; there should be no obstructions between the measured object and the radar water level meter, otherwise it will affect the reflection of radar waves, that is, affect the measurement accuracy; the center of the radar water level meter is far from the shore of the water body. The distance must be greater than the radius of the transmitting beam, otherwise the measurement accuracy will also be affected; the installation cantilever bracket of the radar water level meter must be firm, and cannot be shaken up and down; in order to protect the radar water level meter, the water level meter can be installed on the cantilever bracket. Iron box, put the radar water level meter probe. Discover additional details at radar level measurement.
For the ultrasonic instrument that continuously measures the liquid level, when the temperature of the liquid to be measured changes greatly, the change of the propagation speed of the sound wave should be compensated. The connecting cable between the detector and the converter should take anti-electromagnetic interference measures. The structure of the ultrasonic level sensor should be determined according to the process requirements and on-site working conditions.
In the process of cement production, material level detection and material level alarm control play an important role from mining and crushing to storage and packing of finished cement. But because of the complex working conditions of cement plant, radar level meter has no small difficulty in measuring. Today, I would like to talk with you about specific difficulties and solutions. First cement plant material for measurement in the dust under the condition of particularly large, especially some pneumatic feed bin, involves the kinds of materials, material particle size is differ, dielectric constant is different also, such as broken limestone, raw meal, clinker, fly ash, coal, cement, the level of the bunker, it is very difficult to accurately measure Especially fly ash bin with low dielectric constant.
Measurement accuracy, the accuracy of ordinary radar is generally ±10mm, and the accuracy of precision radar is ±3mm. Selection according to the actual needs of production. Range, according to actual needs, choose the antenna size. Note that the actual range is reduced in complex environments. Antenna type and antenna size, the larger the antenna size, the larger the measured range and the stronger the anti-interference ability. The antenna types of radar level gauge are rod type, bell mouth type, paraboloid and so on.
Mechanical interference, the so-called mechanical interference, means that some components inside the radar level sensor will also vibrate due to the large-scale vibration or impact of the external machine, and even undergo displacement and deformation, which may also cause the pointer of the instrument head to loosen and cause measurement. error. In this case, we usually use spacers, shock springs, etc. to cushion the shock.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. See extra information on https://www.kaidi86.com/. Suitable for chemical industry, petroleum industry, metallurgical industry, water conservancy and electronic industry, etc.
Working principle: Tube in the liquid level meter has a magnetic float, float design, according to the proportion of medium float magnet steel and container liquid level at the same level, wholesale liquid level gauge in a semiconductor magnetel liquid level gauge sensor LED display scale, and orderly arranged from top to bottom, and LED module corresponding to the magnetic field magnetic sensor, the magnetic semiconductor, triggering the corresponding digital circuit, liquid level above the red LED display, The LED below the liquid level gauge is displayed in green, the red in gas phase, and the green in liquid phase. The red-green junction is the actual liquid level in the container.